Original Mani Diamond Burs in Bangladesh
This product is used by connecting to dental handpiece and rotates to grind hard materials and tissues etc..
It is the dental diamond burs with stainless steel shank and non-sterile instrument.
2. Main materials
a) Metal base :Stainless steel (including nickel and chromium)
b) Working part: Fine diamond crystals attached to metal base(including nickel and chromium)
c) Shank : Stainless steel (including nickel and chromium)
MANI coding system reflects shapes of burs. The contour and sizes could be imagined by MANI nomenclature since ISO numbers are also indicated.
●TF/Taper Flat End ●TR/Taper Round End ●TC/Taper Conical End ●FO/Flame Ogival End ●SF/Straight Flat End ●SO/Straight Ogival End ●SR/Straight Round End ●RS/Rounded Shoulder ●EX/Special (Extra) Shape ●BR/Ball Round Type ●BC/Ball Collar Type ●DI/Double Inverted Cone ●SI/Single Inverted Cone ●WR/Wheel Round Edge ●CR/CR Inlay Preparation Dia-Burs ●CD/Children’s Dentistry Dia-Burs
●S/Short Shank ●SS/Super Short Shank ●C/Coarse ●F/Fine ●EF/Extra Fine
Purpose of use and effect-efficacy :
This product has stainless steel working part which is coated with fine diamond crystal. It is defined as the rotary grinding instruments for hard tissues such as tooth and bones. It is used connecting with dental handpiece. It can also be used for grinding metal, plastic, porcelain and other hard materials.
No diamonds palling must be found by a loupe after bending electroplated part at 90 degree.
Method of manipulating or method of use :
According to its intended use, choose the most appropriate MANI DIA-BURS and use it for grinding hard tissues etc.
Possible Combinations with a dental handpiece :
1.Use a dental handpiece which can hold the shank precisely.
(Diameter of Friction grip shank:φ1.6mm)
(Diameter of Contra Angle shank :φ2.35mm)*
2. Use a handpiece which could control at allowable engine speed.
3. Use a handpiece with water spray system.